Do you know the structure of X-ray hand switch? From the outside, the X-ray hand switch is mainly composed of a handle, a base and a spring wire. Regarding the handle, there are the L01 type for domestic conventional use, the L04 type suitable for Europeans and Americans, and the L06 type for dental use. There are holes and no holes for the base. There are three specifications of spring wire of 3 meters, 6 meters and 10 meters. If there are special requirements for the line length, the line can be lengthened after the spring line, but this does not mean that the line is lengthened indefinitely.
Today, a client in a hospital wants to extend the line length of the X-ray hand switch for exposure. It is the hand brake on the mobile bedside machine. The customer means that the X-ray hand switch line is too short and there is radiation too close to the bedside machine. , I want to lengthen the line of the X-ray hand switch. Ask the customer if they want to grow to a few meters, the customer means 100 meters. To be honest, the X-ray hand brake switch that has been sold for so many years, the first time I heard that there is a requirement that the wire length is 100 meters. It’s not that we can’t do it, but if the cable is too long, the sensitivity of the handbrake will be poor. If the customer’s equipment is a mobile bedside machine, the dose is relatively small. If the line is too long and it is far away from the console, it is inconvenient to operate. If you want to reduce the amount of radiation, doctors should wear radiation protective clothing.
Generally speaking, 3 meters, 6 meters and 12 meters of X-ray hand switch cords are sufficient. If you have special needs, please contact us!