Mini toggle switch is controlled for exposure, and the pressing of the first and second gear of x ray hand switch affects the life of the bulb tube. The higher the temperature of the bulb, the shorter the life. When the first leg of the mini toggle switch is pressed, the DR device will immediately heat the bulb filament and immediately lower the filament temperature to protect the filament from exposure to the hot state of about 200 degrees or more.
What is the incorrect exposure of mini toggle switch? Mainly, mini toggle switch is too slow. Some operators are used to put the hand brake band 2 points 2 times according to the first one exposure after press the green circle off, then light up when the second gear when press again, if the second leg of the total pressure time was controlled in about 2 seconds, it is completely correct, however, if the operator in patients with no connection position press a and the second leg, under the condition of the light bulb filament will be under the thermal state what also don’t wait, to shorten the life span of the tube.
Newheek suggests that the correct way to press the mini toggle switch is: once the green circle lights up, you can press the mini toggle switch2 again and again within 1 to 2 seconds to end the exposure and hear the sound of the exposure.
Newheek mini toggle switch is divided into wire and wireless, which adopts mechanical press bar transmission, and the switch module adopts “OMRON” micro switch, with longer service life and higher sensitivity.
Newheek can also provide crystal or air connector.
Our mini toggle switch meets your different needs.