Normally, when you press the first gear of the hand switch, the small green ring in the lower left corner of the display will be extinguished, about 0.5-1 seconds, you can continue to press the second gear, complete the exposure, and hear the prompt “beep”.
The way that affects the service life of the ball tube is the way that the first and second gears of the handbrake are pressed.
From principle, the higher the temperature of the bulb, the shorter the life. When the first gear of the handbrake is pressed, the DR device will immediately heat the bulb filament, heating the filament from a warm state to a hot state of about 2000 degrees. After the exposure, the filament temperature drops immediately to protect the filament.
The correct way to press the brake is: after the exposure green ring is lit, you can quickly, in 1-2 seconds time, one brake two consecutive times, complete the exposure, and can hear the exposure prompt sound.
If you are interested in our hand switch, please contact us email@example.com
Service hotline: +86 536-8882360 / TEL: +86 18953613955
ADD-Factory: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street,
Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China
Copyright 2022 Newheek x ray hand switch and foot switch supplier All Rights Reserved. sitemap
About us: Newheek manufactures different types of X-ray hand switches and X-ray foot switches.
X-ray exposure hand switch is mainly used on portable X-ray, mobile X-ray, stationary X-ray, analog X-ray, digital X-ray, radigraphy X-ray etc X-ray equipments.
As well as X-ray exposure hand switch is applicable to beauty laser device, healthy recovery device etc field.
X-ray foot switch is widely used on medical fluoroscopy X-ray machine, ultrasound, health beauty equipment and industrial etc field.
The quality is assurance, the reasonable price, welcome to order or customize our X-ray hand switch and X-ray foot switch.