Today I will introduce to you the manual of wireless X ray hand switch
One. Technical Parameters:
Transmitter (remote control)
- Remote control transmitter frequency 315-433MHZ (customizable)
- Powered by DC / 9V battery.
- Two-key control, exposure key, reset key.
- The button needs to be pressed for 0.5 seconds before the receiver can receive the correct trigger signal. To prevent misuse.
Receiver (wireless controller)
1: Input voltage: DC 9-20V.
2: Control mode: a. Jog mode b. Delay mode
3: The delay time of the two channels is adjustable from 1-9 seconds;
4: Relay output contacts can be selected as normally open or normally closed, and its output power: 250VAC / 5A 30VDC / 5A;
5: Remote control receiving frequency band 315MHZ or 433MHZ, can be customized;
two. Instructions for use:
- Delay function:
Press the “Exposure” button, the delay starts, the X-ray machine automatic exposure operation is completed within the set time, the exposure is completed, and the buzzer emits an intermittent beep (one time every 1S) that is the end of exposure, press the remote control “Reset” (Sound to stop), the next exposure operation. Press the “Reset” button during the delay to cancel the exposure operation.
Features: Suitable for long-distance exposure control, avoiding X-ray radiation.
- Jog function:
Press and hold the “Exposure” button without releasing it, and complete the X-ray exposure operation within the set time. The buzzer emits an intermittent beep at the end of exposure (one beep every 1S). Press the “Reset” button (tweet to stop) for the next exposure operation. Release the “Exposure” button to cancel the exposure operation at any time.
Features: It is suitable for close-range exposure operation and can end the X-ray machine exposure operation at any time.
If you need to remember to contact us.