I recently received a customer inquiry about the reason why the Minova hand switch was all yellow faults when it was exposed? I am sorry here, because the newheek hand switch has not had any faults. About the Minova hand switch when it was exposed, it was all How the yellow fault is caused, it is best to consult the original manufacturer.
If you want to replace the Minova hand switch, you can choose the newheek hand switch.
L01X-ray handswitch ①Internal structure, double-key double-opening. It adopts mechanical push rod transmission. The switch component adopts “OMRON” micro switch, which has longer service life and higher sensitivity. ②The mechanical life is 1 million times, and the electrical life is 200,000 times
L04X-ray hand switch ①Internal structure, double-key double-opening. It adopts mechanical push rod transmission, and the switch component adopts “OMRON” micro switch, which has longer service life and higher sensitivity. ②50 million times of mechanical life and 300,000 times of electrical life ③The handle is detachable, and customers can assemble by themselves
L06X-ray hand switch ①Used for dental portable machine (single tooth film). 1 gear switch. ②Mechanical life: 1 million times, electrical life: 200,000 times ③2 core 3 meters
L03X-ray hand switch ①3 gears. The green button is the light that controls the beam light. ②The mechanical life is 50 million times, and the electrical life is 300,000. ③6 core 6 meters
X-ray hand switch customization service:
There are 3 meters of 3 cores, 3 meters of 4 cores, 6 meters of 3 cores, 6 meters of 4 cores, 6 meters of 6 cores, 10 meters of 3 cores. The length of the spring wire can be customized according to customer needs. (3 meters refers to the 30 cm spring wire made in a straight line of 3 meters)
If you have any needs, please remember to contact us!