When using the x ray hand switch, pay attention to the speed of the x ray hand switch pressing should not be too slow. Two brakes are pressed twice. After the first brake is pressed, the second brake is pressed again. The pressing time of the two gears is controlled in about 2 seconds, which is the correct operation method.
However, some operators are accustomed to press the first gear, not connected to press the second gear, the bulb filament will wait in the burning state, the filament is seriously damaged, the life of the bulb is shortened.
If the x ray hand switch is stuck, the pressure resistance is large, or the button pops out smoothly after pressing. It will also cause filament loss and shorten tube life. Therefore, to develop a regular check of x ray hand switch, the use of correct methods, once the above adverse situation, need to replace the x ray hand switch.
Service hotline: +86 536-8882360 / TEL: +86 18953613955
ADD-Factory: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street,
Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China
Copyright 2022 Newheek x ray hand switch and foot switch supplier All Rights Reserved. sitemap
About us: Newheek manufactures different types of X-ray hand switches and X-ray foot switches.
X-ray exposure hand switch is mainly used on portable X-ray, mobile X-ray, stationary X-ray, analog X-ray, digital X-ray, radigraphy X-ray etc X-ray equipments.
As well as X-ray exposure hand switch is applicable to beauty laser device, healthy recovery device etc field.
X-ray foot switch is widely used on medical fluoroscopy X-ray machine, ultrasound, health beauty equipment and industrial etc field.
The quality is assurance, the reasonable price, welcome to order or customize our X-ray hand switch and X-ray foot switch.